Revista Medica Herediana
ISSN 1018-130X versión impresa
REY DE CASTRO MUJICA, Jorge, ROSALES MAYOR, Edmundo y FERREYRA PEREYRA, Jenny. Método simplificado para la titulación de la presión del CPAP en pacientes con diagnóstico de Síndrome de Apneas Hipopneas del Sueño (SAHS). : Serie de casos empleando el equipo autoajustable Autoset T.. Rev Med Hered, ene./mar. 2008, vol.19, no.1, p.18-24. ISSN 1018-130X.
The accepted standard method for titration actually accepted is the conventional polysomnography. It consumes time, qualified personnel and has a high operational cost. Objective: To evaluate the auto adjustable CPAP use for CPAP titration in patients with diagnosis of moderate to severe Obstructive Sleep Apnea (OSA). Material and methods: We report 54 patients with the diagnosis of OSA done by polysomnography (6) and respiratory polygraphy (48). We used the Autoset T (Resmed®) autoadjustable system for titration. All patients received during an office interview the instructions about how to use the equipment before they were taken it home. If results of the first test were not satisfactory, the patient would take it home for a second test. Results: 54 studies were done, one for each patient. In the first test, 8 studies (15%) failed, so they took the equipment home once again. Finally, only in 2 patients (3.7%) the titration failed and they needed a conventional titration with polysomnography. The final pressure to be programmed to the CPAP was defined according to the graphic’s visual reading. At the moment we finished the data base, 44 (81%) patients were using the CPAP regularly. Conclusions: We consider that non-conventional titration based on the visual reading of auto adjustable CPAP records is an useful, simple, very cheap CPAP titration method in our country. (Rev Med Hered 2008;19:18-24).
Palabras llave: Obstructive Sleep Apnea; sleep apnea; OSA; CPAP; autoadjustable CPAP; intelligent CPAP; CPAP titration.
Universidad Peruana Cayetano Heredia
Facultad de Medicina "Alberto Hurtado"
Av. Honorio Delgado 430 Urb. Ingenieria, San Martin de Porres
Lima 100 - Peru